The RF field emission probe set consists of a magnetic field probe and an E field probe. They are used to measure respective near fields, which are coupled out by an IC in operation/ a running IC. The measured data help to assess the seperate emission of an IC according to the types of field. Measurements and all calculations are carried out with the software ChipScan-ESA.
Scope of delivery
- 1x P1601, RF Magnetic Field Probe
- 1x P1702, RF E-Field Probe
- 1x CS-ESA set, ChipScan-ESA Software / USB
- 1x BPM 02, dB/dt Field Meter
- 1x EPM 02, dE/dt Field Meter
- 1x D70 h03, Spacer Ring, 3 mm
- 1x D70 h10, Spacer Ring, 10 mm
- 1x SMA-SMA 1 m, SMA-SMA Measuring Cable
- 1x SMA-SMB 1 m, SMA-SMB Measuring Cable
- 1x N-SMA, Adapter
- 1x P1600 / P1700 case, System Case
- 1x P1600 / P1700 m, P1602 / P1702 Set User Manual
Scope of delivery details
The P1601 RF magnetic field probe is used to measure magnetic near fields up to 1 GHz, which are coupled out from an IC. These fields induce a voltage inside a field probe conducter. The induced voltage is measured by a connected spectrum analyzer.
The P1702 RF E-field probe is used to measure electric near fields, which are coupled out by an IC. The fields couple into the near-field electrode. The generated displacement current generates a voltage drop at the input of the connected spectrum analyzer. This voltage drop is measured by the conne…
ChipScan-ESA is a software designed to control a spectrum analyzer remotely. The measurement curves are saved and logged in the software. The user can then combine saved curves with other saved curves, correction curves, frequency-dependant functions (curves), or constants into a single curve. Any n…
The BPM 02 is a dB/dt field meter and it is used to measure the temporal change of the magnetic field strength characteristics dB/dt up to 3 GHz. The lower limit frequency depends on the measurement set-up.
The EPM 02 is suitable for the measurement of fast transients with a bandwidth up to 3 GHz. The dE/dt field meter doesn't have a lower limit frequency. Depending on the noise limit of the used measuring device the usage of the EPM 02 is limited in the lower frequency range. For example, at a noise …